- Weight: 131 gr / 4.6 oz (with battery)
- Length: 95 mm / 3.7 in
- Width: 78 mm / 3.1 in
- Height: 33 mm / 1.3 in
- Volume: 139 cm³ / 8.5 in³
- Includes 1 AAA battery (replaceable)
- Battery life: 6 months (varies with usage & environmental conditions)
- Force Range: 90 kg / 200 lb
- Force Accuracy: within ± (1.4 lb + 5% of the applied force)
- Max load: 110 kg / 250 lb
- System Requirements: Activ5 Training App (free) IOS and Android
- Connectivity Bluetooth: 4.1 (Frequency Range: 2402 – 2480 MHz)
- Operating Distance: up to 2m / 7ft (from the Activ5 app host device)
- Operational Temperature: 0 to 35 °C / 32 to 95 °F
- Storage Temperature: -25 to 50 °C / -13 to 122 °F
- Humidity: 15% – 93% RH
- Pressure: 700 to 1060 hPa (IEC 60601-1-11 clause 4.2.2.)
Safety: IEC 60950-1:2005 + A1:2009 + A2:2013, EU differences to EN 60950-1:2006, USA: UL 60950-1 2nd edition, CAN/CSA
C22.2 No 60950-1-07
cMET Safety mark for USA and Canada, Listing number E114377, certified to UL 60950-1 Safety ITE Part1
FCC Part 15B, 15C; Industry Canada ICES-003 (FCC ID: 2AGCI-A5P1, IC: 22259-A5P1)
FCC RF Exposure Report according to FCC 47 CFR Part 2.1093
EU RF Exposure Report according to IEC 62479:2010
Bluetooth 4.1: FCC Part 15.247, 15.209; CANADA RSS-247 Issue 1, RSS-Gen Issue 4
Bluetooth SIG Qualified design: Declaration ID: D033645
EU RED Directive 2014/53/EU: article 3.1(b): EMC EN 301 489-1 + EN 301 489-17 (also old: RTTE article 3.1b: EN 301 489-1, EN 301 489-17)
EU RED Directive 2014/53/EU: article 3.2: RF Spectrum: ETSI EN 300 328 (also old: RTTE: Article 3.2 per EN 300-328)
Flammability: IEC 60695‐11‐10 HB75 (for external components)
RoHS: Successful tests to meet: Annex II of EU Directive 2011/65/ EU + 2012/50/EU to 2012/51/EU + other EU Directives 2014/ 2015/ 2016/
WEEE: 2002/96/EC and 2012/19/EU + marking standard EN 50419
California Proposition 65
European Union Packaging Directive 94/62/EC and Toxics in Packaging Act USA
Supported Hardware / Operating System
Requires Android 8.0 or later
Requires iOS 11.0 or later (Compatible with iPhone, iPad, and iPod touch)
Software License Agreement
How to recycle your Activ5
Please contact the company you purchased the Activ5 from, or you can contact your local waste management authority.
The Activ5 device should be recycled as “Electronic and Electrical Equipment (EEE) waste”, instead of being disposed with general household waste.
In the European Union, this symbol indicated that this product should not be disposed with other household waste. The device should be deposited at an appropriate facility to enable recycling. Batteries should be discarded separately, at an appropriate collection point.
Activ5 is RoHS compliant.
Please use government or local authority collection points, where you can recycle waste electronic equipment.
You can also contact the store or web site, where you purchased the goods.
Strength Testing Guides
Supported Hardware/ Operating System
Requires Android 8.0 or later
Requires iOS 11.4 or later (Compatible with iPhone, iPad, and iPod touch)
What is the Activforce 2?
The Activforce 2 is a patented dynamometer, inclinometer and easy-to-use companion app that utilizes advanced technology to collect and measure real-time force and range of motion data.
What specifically does the Activforce 2 dynamometer measure?
The Activforce 2 provides accurate measurements of the following:
- Active and Passive Range of Motion (AROM & PROM)
- Peak Force
- Average Force
- Force over Time
- Force/ Weight Ratio
- Muscle Symmetry
- Neuromuscular Deficits
How accurate is the Activforce 2 dynamometer?
The Activforce 2 dynamometer is accurate within 1.4 lbs. +/-% of the applied force.
What is the maximum force the device can measure?
250 lbs. or 110 kg.
What are the technical specifications of the Activforce 2?
The specifications of the Activforce 2 are as follows:
Weight: 137 gr / 4.8 oz (with battery)
Length: 95 mm / 3.7 in
Width: 78 mm / 3.1 in
Height: 33 mm / 1.3 in
Force Range: 90 kg / 200 lb.
Force Accuracy: within ± (1.4 lb. + 5% of the applied force)
Max load: 110 kg / 250 lb.
System Requirements: Activ5 Training App (free) IOS and Android
Connectivity Bluetooth: 4.1 (Frequency Range: 2402 – 2480 MHz)
Is the Activforce 2 dynamometer waterproof?
The device itself and the accessories are water-resistant. They may be wiped down with a damp cloth or disinfecting wipe.
How long will the battery last?
Your Activforce 2 device uses a AAA battery that will last up to a year, depending on usage.
Will my account be tied to a specific device?
To create a new Activforce 2 account, you will need a new Activforce 2 dynamometer that has not been “activated.” Once you create an account, it can be connected to any Activforce 2 device.
Does the Activforce 2 dynamometer ever need to be calibrated or tared?
Your Activforce 2 device should not need to be calibrated. Rarely, you may find that your strength measurements do not seem as accurate. When this happens, go to Settings > Calibrate your device and tare the device.
What comes in the Activforce 2 kit?
In your Activforce 2 kit you will find the Activforce 2 dynamometer with a locking mechanism, a rounded pad, an arched pad for flatter body parts, a leg strap to attach the device to a patient’s leg, a hand strap to attach the device to your hand, a belt to measure pull force and a belt loop plate to secure the hand and leg straps to the dynamometer.
How do I switch out the Activforce 2 pads?
To change out pads, hold the top, softer part of the pad and turn it to the left. The pad should easily twist off. Take the new pad, line it up with the locking mechanism on the device and twist to the right until the pad is secure.
The locking mechanism is designed to hold the pads at multiple angles. Keep turning the attachment at 90 degree angles until it locks into the angle you prefer.
Can I redo a single test within a test series?
After performing a strength or angle measurement test, you will have the option to redo the test. Just tap the “Redo Test” text under the test result screen.
Can I access test data from previous measurement tests I performed in Activforce 2?
No patient data is stored at this time. Once you email a data summary to the email associated with your account or explicitly end a test, the data for the measurements you previously performed is no longer accessible.
Can I store and view patient data later?
No patient data is stored at this time.
Can I create a custom protocol?
Each test series you perform with Activforce 2 is customizable. You can set and store default settings for strength tests, but you cannot create custom protocols at this time.
Are there standard protocols I can use in the new app?
Based on extensive feedback from physical therapists, we have done away with standard protocols in favor of measurement tests that are customized for your patient’s injury.
How do I measure a knee or elbow joint that does not have full extension?
Although the Activforce device tests a range of motion, and does not calculate a starting angle, you can still measure a knee or elbow joint that does not have full extension. See how-to video for step-by-step instruction.
How do I strength test someone who is stronger than I am?
The belt may be used to test a patient or athlete who may be stronger or larger than you.
For more detailed instruction, see How-to video (against wall/ belt)
How can I see a patient’s force-to-weight ratio?
To view force to weight ratio data, tap the “Enter Weight to see F/W %” button on the Strength Summary screen. You will be asked to enter the patient’s weight, then tap “Calculate F/W %.”
How do I change the email associated with my account?
If you need to update the email associated with your account, please contact [email protected]
I own the original Activforce device. Will I still be able to use the old Activforce app?
Yes, we will still continue to support the original Activforce app for use with the Activforce device.
Can I use the new Activforce 2 app with my older Activforce device?
Yes, the new Activforce 2 app works with the original Activforce device, although the Activforce data dashboard is not currently supported in the new app.
Can I use the old Activforce app with a new Activforce 2 device?
No, the new Activforce 2 dynamometer only works with the Activforce 2 app.
Will I be able to activate my data from the original Activforce in the new Activforce 2 app?
Not at this time, but we expect to allow for the migration of legacy Activforce data into the new app in a future version.
How do I activate the new app to work with my Activforce 2 dynamometer?
After you purchase your Activforce 2 device, sign up for an account on the app or log in to an existing one. During this process you will be prompted to activate your device.
How to recycle your Activforce 2
Follow these directions.